Compass Cryogenics Vapor Monitor can be worn as a badge near the breathing zone to measure personal exposure, or placed in a room to measure area concentration.
Occupational safety standards are based on personal exposure monitoring of each employee who may be exposed to the vapor as a regular part of their job. Periodic monitoring of personnel and room concentrations is recommended
The solvent in Collodion is a mixture of Ethyl Ether and Ethyl Alcohol.
|TWA (ppm)||Ethyl Ether 400/Ethyl Alcohol 1000||Ethyl Ether 400/Ethyl Alcohol 1000||N/A|
|STEL/CEIL (ppm)||N/A||Ethyl Ether 500||N/A|
Sampling Medium: Activated High-Purity Carbon Molecular Sieve
Method of Analysis: Gas Chromatography
Reference Method: NIOSH Method 1610
US OSHA Method 7
Recommended Sampling Time: 15 minutes to 8 hours
Minimum Level of Quantitation: Ethyl Ether 0.06 ppm for 8 hours
Ethyl Alcohol 0.2 ppm for 8 hours
Interferences: No interferences are known.
Effect of Temperature: Less than 10% for each 10 °C variation from 24 °C
Effect of Humidity: No effects detected at 30% RH to 92% RH.
Overall System Accuracy:
Storage Conditions: Maximum shelf life one year.
Recommended Holding Time: Two weeks prior to analysis.